This interface is meant to perform basic and advanced reduction of scan data, with the help of a GUI based on the Slice Viewer widget. It was designed for D16, but should work for every instrument producing scan data, as long as it can be reduced by SANSILLParameterScan.
The user can provide data by:
directly providing a NeXus file either by path or through the browse button. This file is then processed through SANSILLParameterScan with default parameters.
reducing a file through the Advanced button, which opens the dialog for SANSILLParameterScan and allows for changing the default parameters.
After the algorithm has run, the output is displayed in a slice viewer-like interface. On the right is a table that shows the coordinates of all the created ROIs and allows their editing.
Some keyboard inputs allow exporting data to workspaces for further analysis (see below).
There 3 modes of interaction with the slice viewer interface :
Clicking and dragging will move the view around.
Clicking and dragging will zoom on the selected region. Right-clicking and dragging will de-zoom. Alternatively, in all modes, using the mouse wheel will also zoom/de-zoom.
Clicking and dragging will define a rectangular region of interest. Multiple such regions can be created, and they are the main analysis tool provided. They can be selected, deselected, moved, and resized by clicking on the control points.
When a ROI is defined, the X and Y axis projections are plotted in the side plots attached to their respective axis. A small red cross is also drawn in the ROI, showing the position of the center of mass of this ROI (assumed to be a good enough approximation of the peak position).
A number of keyboard shortcuts are provided to interact with the data and the ROIs defined.
Deletes the currently selected ROI.
- ‘x’, ‘y’, ‘c’
Export the cuts - the side plots - to workspaces. “x” exports the one along the X-axis, “y” the one along the Y-axis, and “c” along both of them. A different workspace is created for every ROI on each axis, in addition to a workspace containing the side plot itself (which is just the sum of all the other workspaces based on that axis).
Exports the peaks to a table workspace. For each ROI, the detected peak is the center of mass of the ROI. The values exported are the index of the rectangle, the coordinates of the peak in \(2\theta, \Omega\), the data integrated over the ROI, and, if a background has been provided, the integration corrected by the background (i.e. integration of the sample data minus the background data with minimal reduction). NB : this last value will not make sense if the data shown in the data viewer has already been reduced through the Advanced button.
Tries to extrapolate the ROIs currently drawn, with two behaviours:
if there is only one ROI, it is assumed to correspond to the first non-zero theta peak and linearly place other ROIs at regular intervals assuming \(\theta = \Omega\). No ROI is placed at \(\theta = 0\), because it is the beam and does not contain data.
if there are 2 ROIs, it simply linearly interpolates from these two to place more ROIs. Their size is the mean of the original ROIs.
if there are more than 2 ROIs, no linear interpolation can take place and an error message is thrown.